|Minimum Order Quantity||01 Piece|
|Razor||disposable Lieca Pkt of 50 (low & High Profile)|
ADVANCE MANUAL MICROTOME
Storage temperature range - +5 °C to + 55°C
Operating temperature range - +10 °C to + 35
Specimen feed – Horizontal approx. 30mm/1.18 inches, Vertical Approx. 70mm /2.76 inches
Coarse feed wheel must be closer to the user for fatigue free sectioning.
Adjustable hand-wheel balance system to protect the specimen (object head must not drop in to knife when sectioning is stopped)
Precise orientation system: 8° x/y precision orientation system with calibrated controls for quick orientation of specimen for re-cut. System must have provision to rapidly return to exact zero position which is marked by colored indicators.
Hand wheel brake to allow locking of hand wheel in upper position and locking system below to lock in any position.
Section thickness feed to the specimen from 0.5-60 µm with increments of
0-2 µm in 0.5 µm steps
2 - 10 µm in 1 µm steps
10 – 20 µm in 2 µm steps
20-60 µm in 5 µm steps
Relative humidity: maximum 80% non-condensing.
Storage humidity < 80%.
Section thickness range – 1.0 – 60.0 µm
Specimen retraction must be turned off manually approximately in range of 35-50 µm
Vertical stroke: 70-75 mms Maximum cutting range without retraction 69 mm (without specimen orientation 1 µm)
Equipment must have option of horizontal and vertical orientation (Horizontal angle 8°and vertical angle 8°.
Angle of rotation must be +90°.
Trimming stages option of 10 µm and 50 µm.
Mechanical trimming function with trimming lever having 3 notching positions (0, 10 and 30 50 µm)
Repositioning of knife holder base North – South : +25 mm
Maximum specimen size 4-5 cm / 2 inches.
Blade holder for both high profile & low profile with proper base
Universal specimen holder and Wiseman holder must be provided.
Knife holder must not be available in the instrument
10 packets of Low & High Profile blade should be provided free of cost.
Dust cover must be provided.
Demonstration must be shown on demand.
The equipments quoted should be of international standard and must process FDA and / or CE certification.
Service vendor should be available in South India.
List of user from major Government medical institution should be provided.
Warranty for a minimum period of three years.Comprehensive maintenance for 7 years after warranty period. These will be taken into consideration for price bid